Downloads, Information & Videos
– Channels; 4
– Max Voltage; 60 V
– Max Power; 100 mW/channel
– Range; 16 Ω–160 kΩ
– Update Rate; 200 S/s
– Max Error; R < 1 kΩ: 0,1 %
– Max Error; R < 10 kΩ: 1 %
– Max Error; R < 160 kΩ: 10 %
– Enhanced mode: Max Error; 32 Ω < R < 160 kΩ: 0.25 %
FAQ: Can I use ScanEngine for the module?
A: ScanEngine access is not directly accessible for Third-Party modules, but it is quite easy to mix WireFlow C Series modules with other NI modules in a system and still access the NI modules in Scan Mode. Please see application note AB0005-056 AN4 Mixing WF and NI modules for a step-by-step guide how to do this.
FAQ: What kind of connector are used for the module?
A: The WF 3144 and uses a 10-Pos Plug. You should connect a terminal block for 10-Pos Plug to the module. This terminal block is included together with the module when shipped from WireFlow.
FAQ: How many FPGA resources does a C series module consume?
A: The FPGA usage of a specific software and hardware combination is affected by multiple factors such as; FPGA type and size, compilation options, the random starting point for FPGA code placement etc.
Hence, it is impossible to derive an exact figure that defines the FPGA usage of a specific C series module. For customers that are uncertain on whether their application will fit on a specific FPGA with a WireFlow C series module it is recommended to make a test-build. Note that no hardware is needed to make a test build.
For more details, please see AB0005-104 AN16 FPGA Resource Usage for WF C Series modules.
– Four independent, galvanically isolated channels
– Entirely solid-state simulation
– High resolution with non-linear scaling
– Wide resistance range
– On-board calibration memory
– LabVIEW driver included
– VeriStand driver available
FAQ: Does the module work with cDAQ-systems from National Instrument?
A: No. The module is only compatible with the CompactRIO chassis, including the expansions chassis for Ethernet, EtherCAT, R Series and MXI-Express. However, National Instruments has released a CompactRIO series that supports NI-DAQmx, called NI-904x (you can find them HERE). On these chassis you can use all of our C Series modules.
Ecu Test Demo
NI Alliance Partner, WireFlow AB, shows a demo that illustrates how a simple Electronic Control.
– Resistive sensor simulation;
– Pressure sensors, Thermistors, RTDs etc.
– ATE (Automatic Test Equipment)
– HiL testing (Hardware in the Loop)
– Laboratory testing of electronic control units
– Automated calibration and alignment
– Automate manual controls
FAQ: How is the WF 3144 Programmable Resistor Module designed?
A: The module uses an array of real resistors switched with solid state relays to the desired resistance.
Get up and running with the WF 3144 Resistor Module
Magnus from WireFlow shows step-by-step how to get up and running with the programmable resistor for the Compact RIO.
We'd love to hear from you!