Downloads, Information & Videos
LabVIEW Driver for cDAQ
AC0086-005 WF3132 cDAQ support.zip
AC0023-002 WF3132 User Manual.pdf
– Max Voltage 60 VDC/30 V RMS
– Max Current 0.5 A
– Max Power 10 W
– Max Resistance 0.3 Ω
– Max Update rate 1500 S/s
The 32 SPST reed relays can be used in multiple configurations:
• 1 x 32 (1 wire)
• 1 x 16 (2 wire)
• 1 x 8 (4 wire)
• Four banks of 1 x 8 (1 wire)
• 4 x 8 Matrix (1 wire)
FAQ: What kind of connector are used for the module?
A: The WF 3132 uses a standard 37-pin Dsub connector – type Plug. You should connect a 37-pin Dsub connector – type Receptacle to the module. This Receptacle is NOT included with the module when shipped from WireFlow.
– 32 SPST relays
– Multiple configurations possible
– Standard 37-pin Dsub connector
– LabVIEW driver included
– Compatible with NI VeriStand
FAQ: Does the module work with cDAQ-systems from National Instrument?
A: Yes. The WF3132 module works on both CompactDAQ and CompactRIO. Be certain that you select the correct driver for your target device.
FAQ: How many FPGA resources does a C series module consume?
A: The FPGA usage of a specific software and hardware combination is affected by multiple factors such as; FPGA type and size, compilation options, the random starting point for FPGA code placement etc.
Hence, it is impossible to derive an exact figure that defines the FPGA usage of a specific C series module. For customers that are uncertain on whether their application will fit on a specific FPGA with a WireFlow C series module it is recommended to make a test-build. Note that no hardware is needed to make a test build.
For more details, please see AB0005-104 AN16 FPGA Resource Usage for WF C Series modules.
– High channel count test systems
– High channel count control systems
– HiL testing (Hardware in the Loop)
– Production test
– Galvanically isolated interfacing
FAQ: Can I use ScanEngine for the module?
A: ScanEngine access is not directly accessible for Third-Party modules, but it is quite easy to mix WireFlow C Series modules with other NI modules in a system and still access the NI modules in Scan Mode. Please see application note AB0005-056 AN4 Mixing WF and NI modules for a step-by-step guide how to do this.
Ecu Test Demo
NI Alliance Partner, WireFlow AB, shows a demo that illustrates how a simple Electronic Control.
We'd love to hear from you!